In hyperthermia treatment planning (HTP) the goal is to find the amplitudes and phases of antennas in the applicator to efficiently heat the tumor. To do this prior information regarding tumor characteristics such as the size, position and geometry, in addition to an exact model of the hyperthermia applicator is needed. Based on this information, the optimal frequency of operation can be determined. In this paper the optimum frequency for hyperthermia treatment based on the tumor and applicator characteristics, using time reversal as the focusing technique, is studied. As prior information, we consider tumor size and position, the number of the antennas in the applicator and the frequency characteristics. The obtained optimal frequency range is found using hyperthermia quality indicator values calculated from simulations. We also determine the optimum position of the virtual source in the initial step of the time reversal method to increase the quality of the treatment.
2. Datta, N. R., S. G. Ordonez, U. S. Gaipl, M. M. Paulides, H. Crezee, J. Gellermann, D. Marder, E. Puric, and S. Bodis, "Local hyperthermia combined with radiotherapy and-or chemotherapy: Recent advances and promises for the future," Cancer Treat Rev., Vol. 41, No. 9, 742-753, Nov. 2015, doi: 10.1016/j.ctrv.2015.05.009, Epub 2015 May 27.
3. Lee, H. K., et al., "Superficial hyperthermia and irradiation for recurrent breast carcinoma of the chest wall: Prognostic factors in 196 tumors," Int. J. Radiat. Oncol. Biol. Phys., Vol. 40, 365-375, 1998.
4. Franckena, M., et al., "Hyperthermia dose-effect relationship in 420 patients with cervical cancer treated with combined radiotherapy and hyperthermia," Eur. J. Cancer, Vol. 45, No. 11, 1969-1978, Jul. 2009.
5. Turner, P. F., A. Tumeh, and T. Schaefermeyer, "BSD-2000 approach for deep local and regional hyperthermia: Physics and technology," Strahlenther Onkol., Vol. 165, 738-741, 1989.
6. Crezee, J., et al., "Improving locoregional hyperthermia delivery using the 3-D controlled AMC-8 phased array hyperthermia system: A preclinical study," Int. J. Hyperth., Vol. 25, 581-592, 2009.
7. Paulides, M. M., J. F. Bakker, E. Neufeld, J. van der Zee, P. P. Jansen, P. C. Levendag, and G. C. van Rhoon, "The HYPERcollar: A novel applicator for hyperthermia in the head and neck," Int. J. Hyperth., Vol. 23, 567-576, 2007.
8. Togni, P., Z. Rijnen, W. C. Numan, R. F. Verhaart, J. F. Bakker, G. C. van Rhoon, and M. M. Paulides, "Electromagnetic redesign of the HYPERcollar applicator: Toward improved deep local head-and-neck hyperthermia," Phys. Med. Biol., Vol. 58, No. 17, 5997-6009, 2013.
9. Dobsıcek Trefna, H., J. Vrba, and M. Persson, "Time-reversal focusing in microwave hyperthermia for deep-seated tumors," Phys. Med. Biol., Vol. 55, No. 8, Apr. 2010.
10. Converse, M., E. J. Bond, B. D. Van Veen, and S. C. Hagness, "A computational study of ultra-wideband versus narrowband microwave hyperthermia for breast cancer treatment," IEEE Transactions on Microwave Theory and Techniques, Vol. 54, No. 5, 2169-2180, 2006.
11. Dobsıcek Trefna, H., P. Togni, R. Shiee, J. Vrba, and M. Persson, "Design of a wide-band multichannel system for time reversal hyperthermia," Int. J. Hyperthermia., Vol. 28, No. 2, 175, 2012.
12. Dobsıcek Trefna, H., M. Shafiemehr, M. Persson, "Laboratory prototype of UWB applicator for head and neck hyperthermia," 12th International Congress on Hyperthermic Oncology ICHO 2016, New Orle.
13. Paulides, M. M., R. R. Stauffer, E. Neufeld, P. F. Maccarini, A. Kyriakou, R. A. M. Canters, C. J. Diederich, J. F. Bakker, and G. C. Van Rhoo, "Simulation techniques in hyperthermia treatment planning," Int. J. Hyperthermia., Vol. 29, No. 4, 346-357, 2013.
14. Kok, H. P., P. Wust, P. R. Stauffer, F. Bardati, G. C. van Rhoon, and J. Crezee, "Current state of the art of regional hyperthermia treatment planning: A review," Radiat. Oncol., Vol. 10, 196, 2015.
15. CST, , Computer Simulation Technology, Retrieved from https://www.cst.com/Products/CSTMWS.
16. Dobsıcek Trefna, H., M. Shafiemehr, and M. Persson, "Laboratory prototype of UWB applicator for head and neck hyperthermia," 12th International Congress on Hyperthermic Oncology ICHO 2016, New Orleans, USA, Apr. 11–15, 2016.
17. Gabriel, S. R., W. Lau, and C. Gabriel, "The dielectric properties of biological tissues: III. Parametric models for the dielectric spectrum of tissues," Phys. Med. Biol., Vol. 41, 2271-2293, 1996.
18. Lazebnik, M., L. McCartney, D. Popovic, C. B. Watkins, M. J. Lindstrom, J. Harter, S. Sewall, A. Magliocco, J. H. Booske, M. Okoniewski, and S. C. Hagness, "A large-scale study of the ultra-wideband microwave dielectric properties of normal breast tissue obtained from reduction surgeries," Phys. Med. Biol., Vol. 52, 2637-2656, 2007.
19. Canters, R. A. M., P. Wust, J. F. Bakker, and G. C. Van Rhoon, "A literature survey on indicators for characterization and optimization of SAR distributions in deep hyperthermia, a plea for standardization," Int. J. Hyperthe., Vol. 25, No. 7, 593-608, 2009.