There exist some problems that the water content of the test object cannot be reflected in real time. The detection time is too long, and the heating measurement method destroys the seed tissue for the traditional measurement of the water content of the seed. In this paper, a structure of single excitation coil to double receiving coils is proposed to measure the water content of the seed via electromagnetic induction. The relative permittivity of the seed can be obtained by the relationship between the amplitude ratio and the water content of the seed. First of all, according to the electromagnetic field theory, the functional relationship between the amplitude ratio of the electromotive force amplitude signals of the two receiving coils and the water content of the seed is established. Secondly, fifty sets of theoretical values of the mentioned model can be obtained through simulation analysis. Finally, comparative tests are carried out by using soybean seeds. The experimental results preliminarily verify the feasibility of the electromagnetic measurement method of the water content of the seed. The advantage of the proposed method is that the measurement of the water content of the seed is non-contacting.
1. Li, S. X., Y. Y. Gao, Y.-H. Li, and S. X. Tian, "Method of measuring seed water content and its expectation," Seed, Vol. 29, No. 10, 2010.
2. Singh, P., M. Flury, and W. F. Schillinger, "Predicting seed-zone water content for summer fallow in the Inland Pacific Northwest," Soil & Tillage Research, Vol. 115, No. 5, 94-104, USA, 2011. doi:10.1016/j.still.2011.07.005
3. Matthews, S., E. Noli, I. Demir, M. Khajeh-Hosseini, and M. H. Wagner, "Evaluation of seed quality: From physiology to international standardization," Seed Science Research, Vol. 22, No. S1, S69-S73, 2012. doi:10.1017/S0960258511000365
4. Chinese Academy of Sciences, "Chinese Flora 12,", Chinese Flora Editorial Committee, 56–528, Beijing, 2000.
5. Hnijenstei, R. and D. N. NYdam, "Comparison of oven moisture test at 130◦C vs 103◦C," Seed Science and Technology, Vol. 30, No. 3, 102-106, 2002.
6. Ling, J., Z. Teng, H. Lin, and J. Li, "Estimation and fusion method for moisture content detection based on loss on drying method," Chinese Journal of Scientific Instrument, Vol. 39, No. 2, 2018.
7. Chinese Academy of Sciences, "Chinese Flora 24,", Chinese Flora Editorial Committee, 56–528, Beijing, 2000.
8. Chinese Academy of Sciences, "Chinese Flora 12,", Chinese Flora Editorial Committee, 52–86, Beijing, 1988.
9. Fischer, W., S. Beil, and K. D. Krenn, "Karl-Fischer-Reaktion in dimethylsulfoxid," Advanced Synthesis & Catalysis, Vol. 337, No. 1, 266-268, 1995.
10. Du, G. P., Seeds and Seed Physiology, 110-113, Peking University Press, Beijing, 2009.
11. The International Seed Testing Association, International Rules for Seed Testing, China Agriculture Press, Beijing, 2017.
12. Berbert, P. A. and M. Berbertviana, "Meyer and schilz function to estimate common bean seed water content evaluated by radiofrequency," Scientia Agricola, Vol. 64, No. 6, 569-574, 2007. doi:10.1590/S0103-90162007000600002
13. Grosse, C., "A program for the fitting of Debye, Cole-Cole, Cole-Davidson, and Havriliak-Negami dispersions to dielectric data," Journal of Colloid & Interface Science, Vol. 419, No. 4, 102-106, 2014. doi:10.1016/j.jcis.2013.12.031
14. Itolikar, A. B. and M. L. Kurtadikar, "Microwave measurements of dielectric properties of corn vegetation at C-band and comparison with Debye-Cole dual dispersion model," J. Microw. Optoelectron. Electromagn. Appl., Vol. 16, No. 4, 954-965, 2017. doi:10.1590/2179-10742017v16i41087
15. Khamzin, A. A., R. R. Nigmatullin, and I. I. Popov, "Microscopic model of a non-Debye dielectric relaxation: The Cole-Cole law and its generalization," Theoretical & Mathematical Physics, Vol. 173, No. 2, 1604-1619, 2012. doi:10.1007/s11232-012-0135-1
16. Li, Z., J. Y. Zeng, Q. Chen, and H. Y. Bi, "The measurement and model construction of complex permittivity of vegetation," Science China Earth Sciences, Vol. 57, No. 4, 729-740, 2014. doi:10.1007/s11430-013-4691-5
17. Boyarskii, D. A., V. V. Tikhonov, and N. Yu. Komarova, "Model of dielectric constant of bound water in soil for applications of microwave remote sensing," Progress In Electromagnetics Research, Vol. 35, 251-269, 2002. doi:10.2528/PIER01042403
18. Cruciani, S. and V. D. Santis, "Cole-Cole vs Debye models for the assessment of electromagnetic fields inside biological tissues produced by wideband EMF sources," Electromagnetic Compatibility, 685-688, 2012.
19. Liu, X.-F., B.-Z. Wang, and S.-Q. Xiao, "Electromagnetic subsurface detection using subspace signal processing and half-space Dyadic Green’s function," Progress In Electromagnetics Research, Vol. 98, 315-331, 2009. doi:10.2528/PIER09092902
20. Von Brzeski, J. G. and V. von Brzeski, "Topological intensity shifts of electro-magnetic field in lobachevskian spaces. Olbers paradox solved, deep space communication, and the new electromagnetic method of gravitational wave detection," Progress In Electromagnetics Research, Vol. 43, 163-179, 2003. doi:10.2528/PIER03032701
21. Qu, X., Y. Li, G. Fang, and H. Yin, "A portable frequency domain electromagnetic system for shallow metal targets detection," Progress In Electromagnetics Research M, Vol. 53, 167-175, 2017. doi:10.2528/PIERM16111603
22. Peng, Q., Q. H. Li, and Q. Zhang, "Application status of the technology of logging while drilling," Advanced Materials Research, 1010-1012, 1650–1653, 2014.
23. Wei, B. J., G. J. Zhang, and Q. Liu, "Recursive algorithm and accurate computation of dyadic Green ’s functions for stratified uniaxial anisotropic media," Science in China: Series F, Vol. 51, No. 1, 63-80, 2017.
24. Jiang, W., "Study on transmission parameter selection and coil design of multi-frequency induction amplitude and phase logging tool,", University of Electronic Science and Technology of China, 2008.