We present experiments of contact electrocardiograms (ECG) recording using copper and e-textile-based flexible dry electrodes. In this work, dry electrodes with different shapes, sizes, and materials were designed and fabricated. In cardiac monitoring using these flexible dry electrodes, three different conditions were considered, which are sitting, standing, and walking. To evaluate the performances of the fabricated dry electrodes, average-to-variation ratios (AVR) of the recorded ECG signals measured using the flexible dry electrodes were calculated and compared with those measured using the commercially-available wet electrodes in all three conditions. The AVR results demonstrate that the dry electrodes have a similar performance as the commercially-available wet electrodes in the sitting and standing conditions and a better performance in the walking condition. These results suggest that it is possible to weave dry e-textile-based electrodes in normal clothing and use them for continuous monitoring of ECG signals in different conditions.
Mohammad Ranjbar Nikkhah,
"Contact ECG Recording Using Copper and E-Textile Based Flexible Dry Electrodes," Progress In Electromagnetics Research M,
Vol. 101, 47-58, 2021. doi:10.2528/PIERM20092902
1. Searle, A. and L. Kirkup, "A direct comparison of wet, dry and insulating bioelectric recording electrodes," Physiol. Meas., Vol. 20, 271-283, 2000. doi:10.1088/0967-3334/21/2/307
2. Chi, Y. M., T. Jung, and G. Cauwenberghs, "Dry-contact and noncontact biopotential electrodes: Methodological review," IEEE Reviews Biomed. Eng., Vol. 3, 106-119, Feb. 2010. doi:10.1109/RBME.2010.2084078
4. Lopez, A. and P. C. Richardson, "Capacitive electrocardiographic and bioelectric electrodes," IEEE Trans. Biomed. Eng., Vol. 16, No. 1, 99, Jan. 1969. doi:10.1109/TBME.1969.4502613
5. Geddes, L. and M. Valentinuzzi, "Temporal changes in electrode impedance while recording the electrocardiogram with ``dry'' electrodes," Ann. Biomed. Eng., Vol. 1, 356-367, 1973. doi:10.1007/BF02407675
6. Valchinov, E. and N. Pallikarakis, "An active electrode for biopotential recording from small localized bio-sources," BioMed. Eng. OnLine, Vol. 3, No. 1, 25-38, 2004. doi:10.1186/1475-925X-3-25
7. Yan, L. and H. Yoo, "A low-power portable ECG touch sensor with two dry metal contact electrodes," J. Semicond. Technol. Sci., Vol. 10, No. 4, 300-308, Dec. 2010. doi:10.5573/JSTS.2010.10.4.300
8. Yoo, J., L. Yan, S. Lee, H. Kim, B. Kim, and H.-J. Yoo, "An attachable ECG sensor bandage with planar-fashionable circuit board," Proc. Int. Symp. Wearable Computers, 145-146, Sept. 2009.
9. Gruetzmann, A., S. Hansen, and J. Müller, "Novel dry electrodes for ECG monitoring," Physiol. Meas., Vol. 28, No. 11, 1375-1390, 2007. doi:10.1088/0967-3334/28/11/005
10. Tseng, K. C., B. Lin, L. Liao, Y. Wang, and Y. Wang, "Development of a wearable mobile electrocardiogram monitoring system by using novel dry foam electrodes," IEEE Syst. J., Vol. 8, No. 3, 900-906, Sept. 2014. doi:10.1109/JSYST.2013.2260620
11. Gargiulo, G., P. Bifulco, R. Calvo, M. Cesarelli, C. Jin, and A. V. Schaik, "Mobile biomedical sensing with dry electrodes," Int. Conf. Intell. Sensors, Sensor Netw. Inf. Process., 261-266, Dec. 2008.
12. Ishijima, M., "Monitoring of electrocardiograms in bed without utilizing body surface electrodes," IEEE Trans. Biomed. Eng., Vol. 40, No. 6, 593-594, Jun. 1993. doi:10.1109/10.237680
13. Mestrovic, M., R. Helmer, L. Kyratzis, and D. Kumar, "Preliminary study of dry knitted fabric electrodes for physiological monitoring," Proc. 3rd Int. Conf. Intell. Sensors, Sensor Netw. Inf., 601-606, 2007.
14. Yokus, M. A. and J. S. Jur, "Fabric-based wearable dry electrodes for body surface biopotential recording," IEEE Trans. Biomed. Eng., Vol. 63, No. 2, 423-430, Feb. 2016. doi:10.1109/TBME.2015.2462312
15. Choi, Y., J. Lee, and S. Kong, "Driver ECG measuring system with a conductive fabric-based dry electrode," IEEE Access, Vol. 6, 415-427, Feb. 2018. doi:10.1109/ACCESS.2017.2766098
16. Taji, B., S. Shirmohammadi, V. Groza, and I. Batkin, "Impact of skin-electrode interface on electrocardiogram measurements using conductive textile electrodes," IEEE Trans. Instrum. Meas., Vol. 63, No. 6, 1412-1422, Jun. 2014. doi:10.1109/TIM.2013.2289072
17. Oh, T. I., et al. "Nanofiber web textile dry electrodes for long-term biopotential recording," IEEE Trans. Biomed. Circuits Syst., Vol. 7, 204-211, Apr. 2013.
18. Qin, Q., J. Li, S. Yao, C. Liu, H. Huang, and Y. Zhu, "Electrocardiogram of a silver nanowire based dry electrode: quantitative comparison with the standard Ag/AgCl gel electrode," IEEE Access, Vol. 7, No. 2, 204-211, Feb. 2019.
19. Jung, H., et al. "CNT/PDMS composite flexible dry electrodes for long-term ECG monitoring," IEEE Trans. Biomed. Eng., Vol. 59, No. 5, 1472-1479, May 2012. doi:10.1109/TBME.2012.2190288
20. Reyes, B. A., et al. "Novel electrodes for underwater ECG monitoring," IEEE Trans. Biomed. Eng., Vol. 61, No. 6, 1863-1876, Jun. 2014. doi:10.1109/TBME.2014.2309293
21. Luo, C., et al. "An ECG acquisition system prototype design with flexible PDMS dry electrodes and variable transform length DCT-IV based compression algorithm," IEEE Sensors J., Vol. 16, No. 23, 8244-8254, Dec. 2016.
22. Peng, H., J. Liu, Y. Dong, B. Yang, X. Chen, and C. Yang, "Parylene-based flexible dry electrode for bioptential recording," Sensors and Actuators B Chem., Vol. 231, 1-11, Feb. 2016. doi:10.1016/j.snb.2016.02.061
23. Guo, X., et al. "A self-wtting paper electrode for ubiquitous bio-potential monitoring," IEEE Sensors J., Vol. 17, No. 9, 2654-2661, May 2017. doi:10.1109/JSEN.2017.2684825
24. Chlaihawi, A. A., B. B. Narakathu, S. Emamian, B. J. Bazuin, and M. Z. Atashbar, "Development of printed and flexible dry ECG electrodes," Sensing and Bio-Sensing Research, Vol. 20, 9-15, May 2018. doi:10.1016/j.sbsr.2018.05.001
25. Quesnel, P. X., A. D. C Chan, and H. Yang, "Real-time biosignal quality analysis of ambulatory ECG for detection of myocardial ischemia," IEEE Int. Symp. Med. Meas. Appl., 1-5, Gatineau, QC, Canada, May 4-5, 2013.